The 50L Neocision ETL Lab Certified Rotary Evaporator is a full-scale system meant to fulfill the needs of large volume production and meet the requirements of a certified laboratory with its ETL certification standards. Featuring an electric lift bath and all PTFE valves for extreme chemical resistance, the 50L Neocision gives users the ability to recover large amounts of solvent in a smaller footprint with highly improved functionality. All our rotovaps feature high-efficiency valves and drain ports making it easy to directly inject solution continuously without the need to stop the system or release vacuum to drain.
Digital temperature and speed control with a simple user interface
Electric Bath Lift
Teflon Coated Bath for corrosion resistance
High-efficiency drain port and injection valves
Smaller footprint for improved workspace allocation
Mobility with locking casters
Shorter vapor path for faster evaporation and condensation
Direct injection capability for continuous feeding of solution
Continuous draining capability without the need to release the vacuum or stop the system
1-year warranty on all mechanical parts. Does not include glassware/moving parts.
50L Solvent Pro Series Rotary Evaporator
Heater Max Current Draw (Amps)
Optimum Ambient Temperature
5~25C / 41~105.8F
Water Bath Material
304 Stainless Stee
GG-17 High Borosilicate
Water Bath Size
17.7 x 9.8 in.
240V 60Hz Single Phase
Digital Step-less Regulation
190mm / 7.5 in.
Total Power Consumption (kW)
0-99C / 0-210F
49 x 24 x 83 in.
Digital Temperature Control
Maximum Vacuum Pressure
10mm / 3/8 in. Barb
Evaporating Flask Capacity
10mm / 3/8 in. Barb
Receiving Flask Capacity
1 Year - Excluding: Moving Parts/Glassware
Main + Auxiliary Vertical Double Helix
Rotovap Quick Start Guide
This guide gives basic quick-start instructions for rotovaps. It is intended to provide a starting point for learning the process. Parameters should be adjusted to suit specific needs.
Clean and assemble the unit. Make sure all pieces are cleaned and sterilized using an alcohol solution.
Connect the chiller to the condenser and set it to 0C (32F). Always fill a rotovap condenser from the top down (chiller output to top of condenser, chiller input to bottom of condenser).
Connect the vacuum pump to the cold trap and connect the cold trap to a vacuum port on rotovap.
Turn on the heat bath and set it to 40C (113F).
Once the heat bath, chiller, and cold trap have reached desired temperatures, turn on rotovap motor and set the speed to approximately 100 RPM (does not have to be precise). Keep in mind, higher
RPM’s will increase evaporation power but also increase heat load.
Start the vacuum pump and allow the vacuum to pull down for a few minutes before injection.
Once the vacuum has been pulled to a sufficient level (does not have to be precise but should be below 100 Torr), using the injection valve, very slowly inject approximately 500 mL of solution into
the evaporating flask and then close the valve. Allow this small amount to begin evaporating. You may see the chiller begin to rise in temperature, this is called “priming” the rotovap.
Once the temperature stops rising or all liquid is almost completely evaporated, open the injection valve again very slowly so that a small amount of liquid begins to enter the evaporating
flask. Do not close the injection valve this time but continue to allow the solution to enter the evaporating flask.
At this point, you must try to match the input speed to the output speed. What this means is that you should be injecting the same amount of liquid into the rotovap that is equal to the
amount of liquid dripping from the condenser into the receiving flask. You will know that the input speed and output speed are matched because the chiller will remain at a stable
temperature. If the chiller begins to rise, then you are injecting too much liquid. Dial back the valve until the chiller stabilizes.
Once everything is stabilized, you may walk away from the rotovap if necessary. Check every so often to make sure temperatures remain stable and evaporation continues efficiently.
***Note: All units are inspected for broken glass or parts before the unit is shipped. If the unit arrives damaged or glass is broken (this does not include minor imperfections or scuffs in glassware), the Customer MUST contact BVV™ within 3 days of delivery to file a claim and receive a replacement part. If the customer DOES NOT contact BVV™ within 3 days of delivery responsibility for replacement parts falls on the customer and they must pay for replacement pieces.*** ***Note: This item will ship by freight only.***
***Glassware Safety Disclaimer: BVV™ is not responsible for the failure of glassware which must be inspected before and after every use because it may eventually develop imperfections or damage through normal usage, mishandling, and stress caused by temperature variations. If an imperfection or damage is noted while inspecting the glassware DO NOT use the glassware because it can fail to cause bodily harm or damage to the surrounding area.***
50L Neocision ETL Lab Certified Rotary Evaporator
49 x 24 x 83 in.
Let's Grow Together
Sign up for our newsletter to receive exclusive discounts, promotions, and grow tips. We'll also keep you updated on the latest product releases.